- When performing a dry prime, no solvent is pulled through into the priming syringe.
Check valve malfunction.
FIX or WORKAROUND
- Sonicate the check valves.
If sonicating the check valves does not work, replace the check valves. Procedure for replacing check valves is found in the maintenance section of the 2695 Separations Module Operator's Guide