- After processing PDA data with a method set, there are no spectra and only a 2D chromatogram
- The method set does not include "delete PDA 3D channel after extraction"
- Empower 3
- PDA data
The derived channels in the method set are not set up correctly. The channels selected from the drop-down list are the 2D data-collection channels from the instrument method. There are no derived channels created.
FIX or WORKAROUND
Create derived channels for the wavelengths of choice. See KCS 3852 on how to derive a channel in Empower. Reprocess.