- Solvent drops form on the end of ESI probe
- Desolvation heater readback is correct
- Desolvation gas flow readback is correct
- Xevo TQD
Probe tip is bent, causing poor spray.
FIX or WORKAROUND
See KCS article WBK24688 for instructions on how to rebuild the ESI probe.
Electrospray, XEVOTQD, XEVOTQDIVD