- Solvent leaks from the diverter valve
- Leak sensor triggered
- Xevo TQD
- SQ Detector 2
1 - The ESI probe capillary is blocked.
2 - The red PEEK waste line on the diverter valve is blocked.
FIX or WORKAROUND
- Follow the instructional video on how to replace the ESI capillary. View here.
- Reinstall tubing into the fluidics valve port that was leaking.
- Test for functionality.
Replace the red PEEK waste line.