- A "Missing Instrument Configuration" error appears on the MassLynx home page
- The status light is red
- Xevo TQ-XS
An improper reset or a "blip" in the power can cause this issue.
FIX or WORKAROUND
- Close MassLynx.
- Reset the mass spectrometer.
- Open MassLynx and verify that the error no longer appears.
Sometimes the Engineer settings revert to their defaults.
Reload a previously saved Target Registry file.
If a registry file is not available, copy the settings from the experimental file of a sample that was run before this issue arose.