- MS fails source pressure test with an exhaust trip warning
- Xevo TQ
Restricted API gas feed-through line in the source enclosure
FIX or WORKAROUND
- Open the source enclosure door.
- Locate the tubing that feeds the API gas into the source enclosure (feeds through small green gas O-ring/ seal).
- Blow compressed air into the feed line to alleviate the restriction.
- Close the source enclosure door and wait for the source pressure test to successfully complete.