OBJECTIVE or GOAL
Rebooting the EPC on a ZQ or Quattro Micro physically
- ZQ 2000
- ZQ 4000
- Quattro micro API
- Quattro Micro GC
- Locate the unlabeled hole in the front panel, approximately 3/4 of the distance down from the diverter valve to the syringe drive.
- Use PEEK tubing or something similar to push the EPC reset button inside this hole.
- Confirm that the EPC has rebooted successfully by connecting to MassLynx.