- Acquisition stops in UNIFI with "lockmass check" error
- Lockmass intensity is seen at about 1e7
- UNIFI Workstation
- RDa detector
Lockmass signal is too strong.
FIX or WORKAROUND
- Ensure that lockmass concentration is appropriately made in the 2e6 range.
- Restart sample acquisition after confirming appropriate lockmass intensity.