- In UNIFI, instrument setup fails for RDa
- No beam is seen in UNIFI
- RDa detector
- UNIFI workstation
Air in the fluidics lines
FIX or WORKAROUND
- Perform additional purges of the syringe.
- Perform instrument setup once the air is removed from the lines and a beam is seen.