- QDa lost communication several hours earlier
- Restart the system
- QDa displays no beam for resolution and calibration
- QDa displays beam check intensity too low
- Setup configuration includes resolution and calibration
- No other error messages
- "Beam Check Intensity too low" error message when running resolution check
- ACQUITY QDa Mass Detector
- Empower 3
The QDa had an MKII source that requires approximately 30 minutes to fully warm.
FIX or WORKAROUND
- Allow approximately 30 minutes after powering-on.
- Click Resolve to re-run resolution and calibration.