- SQ Detector
- SQ Detector 2
- TQ Detector
The detector requires a supply of dry, oil-free nitrogen with a purity of at least 95%.
The nitrogen must be regulated at 7 bar (100 psi) outlet pressure, using a two-stage gas regulator with an appropriate outlet range, for example, 0 to 8 bar (0 to 116 psi).
Typical nitrogen usage varies from 600-1200 L/h (at atmospheric pressure).