- Accumulator is unable to compress during leak test.
- No solvent comes out of waste line during priming.
- ACQUITY UPLC I-Class.
The intelligent intake valve does not work.
FIX or WORKAROUND
Replace the intelligent intake valve, 700003557.
UPBINARY, UPBSM+, UPIBSM, UPIBSM, UPIBSM+, UPIBSM+, UPISMFL, UPISMFL+, UPISMFTIVD, UPISMFTN, UPISMFTN+, UPPBSM