- All components start as expected when beginning a run in Empower, but there is no signal from the QDa in the real-time plot
- The second injection does not start
The signal connection from the Autosampler is not made.
FIX or WORKAROUND
- In some instances, a start signal I/O cable is needed to initiate a run. Connect the signal cable to positions 5 and 6 on the rear-panel terminal strip.