- When the PDA is powered-on, the initial wavelength verification fails and error message appears in the console
- ACQUITY UPLC PDA Detector
Dirty flow cell
FIX or WORKAROUND
See "Cleaning the flow cell" on page 47 of the ACQUITY UPLC PDA Operator's Overview and Maintenance Guide.