- H-Class SMFTN is set to 5 °C, but it does not cool under 9 °C
- No errors in ACQUITY Console error logs
- Fan in back of SMFTN is running
- Sample Manager, FTN (SMFTN)
Possible condensation or frost buildup
FIX or WORKAROUND
- Set the temperature to 40 °C and leave it overnight.
- Reboot the SMFTN.
- Set the temperature to 5 °C.