- Interpump communications HW error in console
- The error is intermittent; occurs every few hours
- ACQUITY UPLC I-Class BSM
Corrupted ICS and firmware
FIX or WORKAROUND
- Remove the ICS for the BSM and reload the ICS.
- Reflash the firmware on the BSM.
Pump ran overnight with flow from both A and B. Error did not return.