OBJECTIVE or GOAL
Fully reset the Alliance CPU board.
- 2695 Alliance HPLC system
- Power-on the instrument.
- Wait until you hear three beeps or see the Alliance logo on the LCD.
- Press Clear, and then press 0 on the keypad.
- The display states that it is going to factory defaults and to power-down the instrument.
Clear 0 function clears all methods, the seal pack calibration, the configuration, and all logs, including the Trace log.