- The HDX-2 automation PAL arm hangs over the injection port, and the injection is never triggered
- All the inject start cables are in the correct position
- The event triggers are on in MassLynx
- All positions are taught on the CTC-PAL3
- MassLynx 4.1
- Xevo G2-S
The inject start command in MassLynx is different than the command in Chronos .cam method.
FIX or WORKAROUND
Use the attached HDx_Waters_Online_Digestion_Xevo_G2_S.cam. This contains the proper code Key Value and MassLynxStatus for step 95.
The online digestion method is written for SYNAPT G2-Si. The inject start command is "waiting for injection". The inject start command for a Xevo G2-S is "Waiting for inlet start 1". This is changed in the Key Value and MassLynxStatus lines for line 95 in the online digestion method.