- Baseline is very slowly drifting down
- Instrument was powered-on overnight
- Sample and reference flow cell thoroughly purged
- LED optimized
- New mobile phase
- Very low pump pressure ripple
- New assay
- Flow rate accurate
Temperature was set on the column, but not on the detector.
FIX or WORKAROUND
- Set detector temperature.
eluent, pressure fluctuation