- 2489 shows "yes" for status ok in the node properties
- 2489 instrument control panel display works, lamp icon is green
- 2489 stays in a ready state when injection occurs
- 2489 UV/Visible Detector
- 1525 Binary HPLC pump with manual injector
- Empower 3 stand-alone
I/O connections in the wrong port.
FIX or WORKAROUND
- Inject start connections go in the II connection in the rear of the 2489.