- No concentration results reported in TargetLynx
- Peaks are properly integrated
- Found: Response value reported a 0.00 for all peaks
- Found: Response type set to "Internal (relative - use internal standard)"
- Found: Concentration of Standard: Level set to "Fixed", with Concentration of Standard set to 0.00 for the Internal Standard compounds
Concentration of Standard for the Internal Standard compounds is set to 0.00.
FIX or WORKAROUND
- Set the Concentration of Standard for the Internal Standard compounds to either the exact concentration or the value 1.00.