- The calibration check report shows failure
- MassLynx 4.1
- Xevo G2-XS
The calibration check acceptance criteria are not met.
FIX or WORKAROUND
- Calibrate the instrument.
Note: The threshold to define whether a calibration is acceptable can be defined by the user. The acceptance criteria can be set to ppm or Da within IntelliStart's Check Calibration routine.
Note: A calibration check is not expected to match the pass criteria of the calibration due to the use of an external lockmass (even after just calibrating).