- Start a sample set in MassLynx, but the ACQUITY does not inject
- The MS goes straight to acquiring
- LCMS acquisitions from the sample list
The events and triggering is not turned on.
FIX or WORKAROUND
- Go to the inlet editor and select the Tools drop-down menu.
- Select Instrument Configuration.
- Select Events and Triggering in the bottom-right corner.
- Select Events In 1 and next.
- For the pump, it is okay to leave it as Trigger by contact closure (even when not using a contact closure cable).
- Select Next, Finish, and Finish.