- Inlet does not inject
- MassLynx with any mass spectrometer
Triggering was disabled to perform the PM.
FIX or WORKAROUND
- Go to the inlet method and select Tools > Instrument Configuration > Events and Triggering. Select the "Event In 1" check box.
- Reboot the system: first the PC, then the inlet, then the Mass Spec (EPC).