- Sample is not injected
- Mass spectrometer goes straight to acquisition instead of waiting for inject start
- Sample list needs to be aborted to get first line of sample list to stop running
- The events/triggering standard configuration is not working correctly.
FIX or WORKAROUND
- Go to Inlet Editor > Tools > Instrument Configuration > Events & Triggering.
- Change the normal configuration of the events/triggering setting from event in 1 to event in 2.
- Alternatively, change the Trigger options (can be "Trigger by Software" or "Trigger by Contact Closure", depending on the instrument).