- Empower sample sets stop in the middle of acquisition and freeze.
- On one acquisition client, two systems are running, each with a PDA, and one fail.
- Previously collecting 3D scans at 10 points/second and data rate was changed to 2 points/second.
- No errors in the message center.
- No evidence of buffering in the recover log.
- There is enough free tablespace.
- No errors in the recover log.
- Empower 3 FR2
- 2998 Photodiode Array Detector (PDA)
The virtual memory on the acquisition client was not set to 4x the RAM per the installation guide.
FIX or WORKAROUND
- Adjust the virtual memory to 4x the RAM on the acquisition client per the installation guide.
- Restart the client.