- 2695 Alliance will not inject
- Instrument failure @ IEEE (xx) observed
- BusLace status shows "OK" in Node Properties
- All IEEE instruments show OK? "NO" in the node properties
- Ethernet system connects without issue
- Empower Enterprise
- Waters Alliance HPLC
- Alliance Separations Module 2690/2695/e2695
- 2487 Dual Wavelength Absorbance Detector
Instruments were powered-on after the LAC/E.
FIX or WORKAROUND
- Reboot the LAC/E.