During an acquisition, when a user selects "abort and continue to next line", Empower does not continue to the next injection, and an instrument failure message appears
Message in the message center and the diagnostic log: "Starting analysis aborted via CAN"
- Empower 3
- Agilent ICF Support 3.0
- Agilent ICF Support 3.1
- Agilent ICF A02.04
Unknown. This issue was opened as defect PCS 56518.
FIX or WORKAROUND
This issue was fixed in ICF v3.2.