- GC with Headspace was just qualified
- Instrument failure observed when trying to set up an injection
- No other errors are observed
- Agilent 7890 GC
- Agilent Headspace 7697A
- Agilent Instrument Control Framework (ICF) v3.0
The GC configuration changed.
FIX or WORKAROUND
- With ICF v3.0, you must scan and save the configuration of the GC when the configuration of the instrument changes.