- GC with Headspace was just qualified
- Instrument failure observed when trying to set up an injection
- No other errors are observed
- Agilent 7890 GC
- Agilent Headspace 7697A
- Agilent Instrument Control Framework (ICF) v3.0
The GC configuration changed.
FIX or WORKAROUND
With ICF v3.0, you must scan and save the configuration of the GC when the configuration of the instrument changes.