- Acquire PDA and MS data
- Run and report data
- Report displays error message for PDA data, and no plot appears
- MS spectrum index report is OK
- ACQUITY PDA
- ACQUITY TQD
The PDA data acquired was single-wavelength 2D data; the spectrum index report uses 3D data.
FIX or WORKAROUND
- Edit the instrument method to acquire 3D data.
- Run and report.