- "Please enter a value between 0.0 and 10.0 for this item" message observed when entering 20 µL for injection volume
- ACQUITY UPLC H-Class
- ACQUITY UPLC Sample Manager-FTN with 50 µL extension loop
Configuration set incorrectly.
FIX or WORKAROUND
- After installing the desired extension loop, you must modify the loop volume configuration.
In the ACQUITY UPLC Console, select Sample Manager - FTN from the system tree, and then select Configure > Volumes. Select the appropriate extension loop size from the list in the dialog box, and then click OK.