- Run Samples window freezes when Equilibrate function is used
- When this occurs, must restart LAC/E to regain control
- Agilent LC 1100 system with valve(s) like G1158, G1159 or G1160
- Empower 3 FR2
- Agilent ICF 2.2 with ICF A.02.04
- FusionQdB v9.8 SR3
Use of valve or valves with Agilent LC causes Equilibrate issue. When valves are removed, Equilibrate runs okay.
FIX or WORKAROUND
Use Condition Column, instead of Equilibrate, at this time.
Reported as PCS 56154.
Presence of FusionQdB does not matter.
Listed as a known issue with Agilent ICF 3.0.