- An active sequence is running when the user aborts the run
- User makes an adjustment and starts the run again
- Then a "Server Busy" error appears and Empower freezes. The only remedy is to reboot the LAC/E.
- Empower 3 FR3
- Agilent ICF Support v2.2
Defect PCS 55979
FIX or WORKAROUND
This defect is fixed in Agilent Instrument Control Framework (ICF) Support Version 3.0. Please upgrade as described in the Release Notes.
EMP2LIC, EMP2OPT, EMP2SW, EMP3GC, EMP3LIC, EMP3OPT, EMP3SW, EMPGC, EMPGPC, EMPLIC, EMPOWER2, EMPOWER3, EMPSW, SUP