- Sample Set pauses at the Purge Detector line and does not proceed to the next line in the Sample Set
- Empower 3 SR 2
- 2410 RI detector
- e2695 Alliance
PCS 51061 - For systems that include a 2414 detector, when you initiate a sample set that contains a Purge Detector function, and then close the Run Samples window, the sample set pauses after the Purge Detector function completes (see http://www.waters.com/waters/support.htm?lid=51061&type=DFCT)
FIX or WORKAROUND
The Sample Set run resumes when you reopen Run Samples. Alternatively, to avoid this issue, do not close the Run Samples window when you have a Purge Detector line in the Sample Set.