- Sample Set pauses at the Purge Detector line and does not proceed to the next line in the Sample Set
- Empower 3
- 2414 RI detector
- e2695 Alliance
PCS 51061 - For systems that include a 2414 detector, when you initiate a sample set that contains a Purge Detector function, and then close the Run Samples window, the sample set pauses after the Purge Detector function completes (see PCS ) 51061
FIX or WORKAROUND
The Sample Set run resumes when you reopen Run Samples. Alternatively, to avoid this issue, do not close the Run Samples window when you have a Purge Detector line in the Sample Set.
If possible , configure your system to include only the e2695 and 2414 detector
This is related to the firmware version.
- With 2.04 once under Empower control the 2414 switches the mode to 410.
- With 2.10 with the 2414 ICS loaded this does not happen, it stays in RIU.
Same issue if instead of Purge Detector there is a Purge Injector function.
2414, 2414B, 2690, 2690D, 2690DUPGD, 2690UP, 2695, 2695D, 2695DE, 2695DEB, 2695DUP, 2695E, 2695EB, 2695INERT, 2695UP, ALLCOLCLR, ALLCOLHTR, ALLCOLHTRB, EMP2LIC, EMP2OPT, EMP2SW, EMP3GC, EMP3LIC, EMP3OPT, EMP3SW, EMPGC, EMPGPC, EMPLIC, EMPOWER2, EMPOWER3, EMPSW, SUP