- Extracted oil leaks from coupler
- See attached photos
- Biobotanical Extraction System
- SFE 2x5L
- SFE 2x10L
The seal and/or the O-ring in the coupler has failed.
FIX or WORKAROUND
- Replace the O-ring and/or the coupler.
- ASSY, Coupler, SS, 1/4 NPT, 10K PSI p/n 700011052
- O-ring, #013 BUNA-N 90 p/n 700010762
Find additional part numbers for the quick-connect fittings here.