- When starting the LC, the HDX gradient remains at initial conditions and does not start
- Gradient starts running for about five seconds but then returns to initial conditions
- HDX and HDX2 System
- nanoACQUITY UPLC
- ACQUITY UPLC BSM
- M-Class Binary Solvent Manager (uBSM)
Inlet configuration contact closure is set to trigger by the software.
FIX or WORKAROUND
When configuring the inlet,ensure that the detector is set to trigger on contact closure for both the pump and the detector.