- Console shows "Source Door Open" error message
- The source door is closed
- Xevo TQ-S
Poor connection of the source enclosure cable
FIX or WORKAROUND
- Disconnect the source enclosure cable from the instrument chassis.
- Reconnect the cable into its socket.
- Ensure that the error message has cleared and that the Console status is green.