- Cannot acquire CCS data
- Can acquire Tof MS data
The CCS calibrations are not calibrated and have been disabled in Quartz.
FIX or WORKAROUND
- Have Waters rep log in to Quartz and ensure the CCS calibration is enabled
- Put Major Mix on C and Leucine Enkephalin on B
- Run instrument setup and complete the CCS calibration
- Acquire CCS data in analysis