- MS peak area increasing injection to injection over run.
- Method dependent.
- All ACQUITY diagnostics pass.
- Refreshing sample, mobile phase, and column did not resolve issue.
- XEVO TQ MS
- Xevo TQD
- Xevo TQ-S
Charging effect caused by a dirty ion block and guide.
FIX or WORKAROUND
- Clean ion block and guide following the appropriate Overview and Maintenance Guide.
A dirty source enclosure can also cause this charging effect, so if cleaning the ion block, clean the source enclosure as well.