- Xevo TQ-S micro shows the error "Desolvation temperature failed to settle".
- Source recognition: "unknown 0"
- Xevo TQ-S micro
The probe is not properly seated.
FIX or WORKAROUND
Reseat the probe.
After reseating, source recognition is 1.9 V and desolvation temperature is reached correctly.