- "Power supply 15V out of range" error upon powerup in the console under BSM (pump)
- Reset and reboot of the BSM does not resolve error
- ACQUITY UPLC
Bad power supply
FIX or WORKAROUND
Contact Waters for on site service.
Power supply part number 700002592 (If Power Supply doesn't fix it, it could be the PCB or one of the fuses F2 and F3)