- BSM is unable to compress on accumulator pump during leak test
- ACQUITY BSM
A fitting is leaking on the steel tubing between the pump head and the pressure transducer.
FIX or WORKAROUND
Replace the steel tubing (430001120).
The transducer to check valve tubing is also a possible cause and may need replacing (I-Class: 430002583, Classic Acquity i2V: 430001773, Classic Acquity Passive: 430001121).
checkvalve, UPBINARY, UPBSM+, UPIBSM, UPIBSM+, UPPBSM