- Newly created Agilent LC system in Empower
- "Thermostat Installed" and "Thermostat Controlled by Method" are selected in the Pre-Configuration Utility for the Autosampler
- Newly created instrument method
- Empower 3
- Agilent Instrument Control Framework (ICF) Support Version 3.0
Temperature mode is not set in the control panel on the Run Samples screen, as instructed in TECN134909208.
FIX or WORKAROUND
Set the Temperature Mode to Variable in the control panel for the Autosampler.
Using the Agilent G1330A/B Autosampler Thermostat with Empower - TECN134909208