- Peak area varies from injection to injection, high % RSD, poor reproducibility.
- Peak shape inconsistent at apex
- 2424 ELS Detector
Nebulizer is going bad, giving poor spray.
FIX or WORKAROUND
Replace the nebulizer (205001124 - Standard High-Flow Nebulizer).
If the system is qualified, replacement of the nebulizer requires ELSD requalification of the detector tests.