"Organizer configuration modified" error seen for ACQUITY UPLC Sample Organizer
Rebooting PC and modules does not resolve issue
- ACQUITY UPLC
The sample organizer detected a change in configuration.
FIX or WORKAROUND
Remove all sample plates from the sample organizer.
In the ACQUITY console, go to Sample Organizer > Configure > Scan and Store Shelf Layout, and perform the function twice.
Return the sample plates to the sample organizer and click the "Verify" button in the Sample Organizer console.