- No peaks observed following injection of sample on a 2795
- Alliance 2795
Deck offset calibration is off.
FIX or WORKAROUND
- In the instrument console, select "Diag".
- Select "Change needle". (The needle moves to the back of the XYZ assembly.)
- Press "Ok". (The needle moves back to the injector assembly.)
- The needle deck offset calibrates automatically.