- When using Bar as the unit of measurement, if a user sets a flow rate (µL/min) on the µBSM and a high pressure limit in Bar (e.g., 600), starting the flow leads to immediate overpressure as the high pressure limit changes to a very low value (e.g., 50 bar)
- If the high pressure limit is set in psi or kPa, the µBSM functions normally
- ACQUITY M-Class
Unknown software bug
FIX or WORKAROUND
None at this time. The bug is reported as a PFD. A workaround is to use psi or kPa as the unit of measure.