In the console for the BSM, in the square called "Conditions", the pressure is fixed at 15000 psi; A1 says 10000%; Vent valve: Vent and degasser 0.00 psi; BSM Total volume pumped 0.000 L
No readbacks in the Additional status square in the Inlet window; sometimes it says "Not Connected" in this square
Not possible to enter a max. pressure value higher than 15000 in the inlet method
- ACQUITY UPLC I-Class BSM
- ACQUITY Driver 2018, SR1 (ICOP 1.71)
Corrupt ACQUITY driver
FIX or WORKAROUND
Uninstall the driver for the ACQUITY BSM, and then reinstall the same version.