- Communication error for ACQUITY PDA only
- Other components in the ACQUITY system are okay
- Empower 3
- ACQUITY UPLC PDA Detector
- 2998 Photodiode Array Detector
Timing issue between PDA and node
FIX or WORKAROUND
- Power-off the PDA.
- Reboot the node that controls the PDA and let it initialize.
- Power-on the PDA and let it initialize.
- Reboot the node that controls the PDA, again, and let it initialize.
- Open a Run Samples window for the system, and then open the ACQUITY console.
- Reset the PDA via the console.