- Injection set up without error
- Manual injection was made, and injector returned to load position
- Data acquisition does not begin
- No errors observed
- All components show status ok "yes" in node properties
- Reboot of PC and system does not resolve
- Breeze 2
- 1515 Isocratic HPLC Pump
- Manual injector
- 2489 UV/Visible Detector
The inject start cable is disconnected.
FIX or WORKAROUND
- Reconnect the inject start cable.